jose hernandez EDGAR hernandez
"xray firing order is out ouf sequence"
August 21, 2019 09:34
Error generating initial calibration results in the following error: "xray firing order is out ouf sequence. State machine error in task 0 index 305."
The calibrations of the table (motors), also of the openings have already been made. When generating the tests with the shot, the graph increases the value of green color, but the values of brown color do not change.
to rate and post a comment
re: xray firing order is out ouf sequence
August 27, 2019 09:07
It seems that you have a problem ether in the filter drum or the generator, try to do the drum alignment and filter belt should be done correctly.
Are you looking to Buy or Sell a Laser? Do you need Service, Repair or Parts for you Laser or IPL system? Call LASER-TECH today at 855-527-3700 We Sell and Service all major brands and manufactures!