August 02, 2016 06:14
We have some problem with our Senographe DMR.
When we switch ON the circuit breaker, the STOP button is lighting, and when we switch ON from the Panel, there is nothing appear on the Display and we have Red LED lighting in 200-PL2 Supply Command Board.
Appreciate your help.
My e-mail: firstname.lastname@example.org
to rate and post a comment
re: Senographe DMR
August 03, 2016 06:01
I would first check the line voltage and resistance
August 04, 2016 11:00
what LED is on?
re: re: Senographe DMR
May 29, 2020 07:43
juan pablo amado
re: re: re: Senographe DMR
May 29, 2020 08:13
Check 800-PL1 Gantry Power Board F10
Problem of the power supply
August 04, 2016 11:22
check lambda power supply (400 T1) back side of 400 PL1; for +5 +12 and -12 Vdc
August 04, 2016 11:23
Are you sure there is not an emergency button pressed?
ElsMed engineers are located in four different countries. 24 hour response time to any service call. Service agreement provided for end-users range from a 6 month period to a 3 year period and covers all system spare parts.