rf-power in mri
February 03, 2018 11:47
MRI magnetum simense has incons. rf-power during ADJ/TRA.
Do I need to caliberate RF with body coil.
can someone let me know please what to do?
to rate and post a comment
re: rf-power in mri
February 04, 2018 09:27
What MRI Siemens . Body coil is inside you need in one case dummy load and big loader with phantom in service mode in others cases you do not need a dummy load
February 05, 2018 09:49
Please mention which model and trouble shooting depends on when this error came . During installation? While machine running and its a break down ? sajubhas@Gmail. Com
February 10, 2018 12:05
Please share error logs and model for support mail me at firstname.lastname@example.org
October 21, 2019 10:09
do RF characteristic test center the phantom and from local service apply this Tune Up >>> RF characteristic if still doesn't fixed inform me here and your device model too
October 21, 2019 10:10
Injector Support & Service (ISS) offers telephone technical support,
depot repair services, loaner injectors, parts identification and sales, PM tools
and PM parts kits for medical contrast injectors.