Emmanuel Akinyemi Adebisi
Planmeca promax x and y collimator calibration
July 30, 2020 10:46
Any time exposois made the machine comes with error 205While the system monitor displayed grap image too highNeed assistance to resolve this error
to rate and post a comment
SPECT, PET, CT and PET/CT equipment sales and service from OEM certified FSEs. Maintenance, repair, installation, relocation, and parts available with 24/7 support. Call 1-886-NUC-MED1 or email sales@TTGimagingsolutions.com.