September 14, 2015 08:34
My CS 8000C is not capturing image. Once I pressed the exposure switch the message pop up saying " Problem with image reception".
I replaced the digital sensor, the CJ653 board and the CJ664 but but still the same no image captured, I got the same massage popping up. Error24.
to rate and post a comment
re: Kodak 8000C
September 14, 2015 06:13
are you sure it's E24? try with different PC, for start. Laptop... contact me at email@example.com
September 15, 2015 08:07
E24: Heating Voltage Self-calibration Time-out ?
September 15, 2015 08:09
Your problem is with pan images or ceph images?
Please specify the model of your 8000: connected with PCI board or with Ethernet for the image acquisition?
Did you check the alignment of the X-ray beam?
re: re: Kodak 8000C
September 16, 2015 08:00
Both pan and ceph image problem. No any image coming out during and after the exposure.
September 16, 2015 08:02
My problem is both pan and ceph. no image appearing on my acquisition window. It was connected thru ethernet. I tried it with my laptop thinking it might be a software issue but still the same.
re: re: re: Kodak 8000C
September 16, 2015 04:24
In case the unit is not giving errors from the X-ray side, try the following:
-using a long enough ethernet cable connected directly to the pan sensor and to the PC,try to make image. This is to bypass any ethernet cable, or coupler problems.
(The internal ethernet cable inside the column can be damaged due to stress movements)
-The ceph control board can be defective as it carries the pan image and the ceph image to the PC. (If you get a pan image with the ethernet cable connected to the pan sensor, then connect back the pan sensor as before and connect your ethernet cable to the output of the ceph control board and try to get both pan and ceph images)
-Check the DC voltages of both sensor power supplys according to the service manual.
Check also TCP/IP protocol for the correct IP address and Subnet mask.
The firewall can also block the images. Check the settings.
re: re: re: re: Kodak 8000C
September 23, 2015 08:24
I do the step one by one like: by-passed the ethernet cable and connected it directly from pc to the digital sensor, I replace the power supply of both sensor and CJ664 board but still no image appeared in the acquisition window.
The TCP/IP was set as automatically obtain IP address.
re: re: re: re: re: Kodak 8000C
September 24, 2015 07:12
Did you check that you are not in demonstration mode?
re: re: re: re: re: re: Kodak 8000C
September 24, 2015 05:47
Even in a demonstration mode no dark or demo image appeared. I can only see the timer countdown and tube head cooling during and after exposure on the acquisition window. It is nothing but an acquisition window on the screen. Firewalls and antivirus are all off.
re: re: re: re: re: re: re: Kodak 8000C
September 25, 2015 08:02
Demo mode is a feature that prevents radiation during trade shows. The unit acts as if it was making X-rays and no error message apears. The image shown is generated by the software, not by the unit.
Help from someone still working at Kodak would be appreciated in order to follow the problem since it appeared(just after installation or later. Also if you had a training course on Kodak products, you may have some contacts with other people attending the same course as you.
re: re: re: re: re: re: re: re: Kodak 8000C
September 25, 2015 09:23
This problem can't be solve by Carestream third party service provider. Even he install new KDIS and new driver, updated the masking files but still the same. I am not disregarding the software issue even I used three different pc. What I am thinking now is to replace the rotating arm assy instead of checking the cable one by one as it may disrupt the exposure. I'll keep you posted.
September 27, 2015 02:51
do you still have the service manual for this unit if so, could you email it to me at: firstname.lastname@example.org - thanks.
re: re: re: re: re: re: re: re: re: Kodak 8000C
May 15, 2018 12:03
May I also have a copy of the service manual. power board issues. Wayne
re: re: re: re: re: re: re: re: re: re: Kodak 8000C
October 22, 2018 02:48
May I also have a copy of the service manual of kodack power board issues8000C. Thank a lot
re: re: re: re: re: re: re: re: re: re: re: Kodak 8000C
October 23, 2018 09:04
Link sent by mail, thanks
March 08, 2019 10:28
can you send me the service manual of kodak 8000C? I really need your help,i can not reset the arm position. it always tells wrong arm position.my email is email@example.com
March 09, 2019 02:43
Link to service manual sent by email to you.
August 06, 2020 12:32
I am not able to raise and lower the kodak 8000C for some reason. When I press the buttons to raise and lower, it just makes a clicking sound. Any potential help or advice of how to fix it?
February 12, 2018 09:16
Can you send me the service manual please? Hadrieloextr@hotmail.com
February 12, 2018 05:10
Ok, done, Thks
Hello. Can you point me to finding a copy of the Kodak 8000 service manual? I would greatly appreciate it.
May 15, 2018 12:06
Hello. I am in search for the service manual as well. Can you help?
June 17, 2019 10:18
I have been looking for the service manual but can not find it. can some one please share a link
June 17, 2019 05:10
Link sent trough Dotmed message center as you didn't mention any email,
June 18, 2019 09:48
i got it thank you so much!
Adepto Medical is an infusion pump rental, sales, and service company. Our focus is to deliver to our customers the best product and customer service support we can offer. Call (913) 261-9933