Kodak 8000 Pano Carestream Error/Calibaration
April 20, 2020 10:12
I have Kodak 8000 Carestream Pano, whenever turned on always shows different errors- sometimes shows E15, E53, or cooling tube although we do not use radiation.
Please help. Thank you
to rate and post a comment
re: Kodak 8000 Pano Carestream Error/Calibaration
April 21, 2020 10:16
Change the RTC (Real Time Clock on CPU board. It can be a
M48T86 thick black IC or an yellow battery inserted on an IC.
After replacement you get err 58, then cacel the error and make a full pan. Make sure you insert the components in the right direction. Thanks for reading
re: re: Kodak 8000 Pano Carestream Error/Calibaration
April 22, 2020 10:27
I did change the battery as it was not able to take any pano. Since then it’s giving the different errors mentioned above.
It has yellow battery. I do not know may be new battery is also expired( bought from amazon) . Do you have any suggestion, from where I can buy one?
June 02, 2020 11:13
I also have the E53 problem, can you send me the manual to fix it?
re: re: re: Kodak 8000 Pano Carestream Error/Calibaration
June 03, 2020 09:42
Hi, The solution has been sent by email to you.
Thanks for reading,
Call Master Medical today for a great value on our large in-stock selection of
Defibrillators including Zoll, Physio Control, Philips with our Low Price Guarantee and Quick shipment. (866)468-9558.