GE Revolution ACT 16 slices problem
October 22, 2019 10:41
Hi to all mates, i am new to this forum.
I am facing problem with Ge revolution ACT 16 slice CT, when daily warm-up is launched its aborted by CT, and the following errors displayed in errors journal.
Sat oct 19 16:48:32 2019
Host: tgp proc: tgp_subsys Error: 260100032
File: scTrigSeqActions. line: 1986
Sat Oct 19 16:48:32 2019
Host: Table Gantry Processor Ermes # : 260100032
Exception Class : Abort Severity: Primary
File : scTrigSeqActions.c Line# : 1986
Function : XRAY GENERATOR : Final Exposure Command
Scan Type : Axial Scan : 50272/1/2
Timeout on Exposure Command reset to X-ray off!
HW Error : Possible hardware problem.
Timeout is : 300 mS.
(KV: 80 mA: 20 GenOpt: 0x8 Freq: 984 Ax: 1000 Dur: 992).
Any help or support will be appreciated
Thanks in advance.
to rate and post a comment
re: GE Revolution ACT 16 slices problem
October 23, 2019 10:25
Timeout on exposure command is the system has not been given command to expose. Possible issue with GUI or staff not hitting scan button in time and scan aborting?
Our engineers are trained strictly on Siemens & GE PET to Guarantee an expert on all service calls. Our large inventory of PET & PET/CT systems & parts
allows us to provide a TRUE Turn-key solution. Ph:866-645-7457