Carestream Classic CR problem
May 11, 2020 10:40
Classic CR scanner not extracting the plate from chassis.
to rate and post a comment
juan pablo amado
re: Carestream Classic CR problem
May 15, 2020 11:42
do you have error code?
re: re: Carestream Classic CR problem
May 25, 2020 11:04
re: re: re: Carestream Classic CR problem
May 26, 2020 10:47
I can´t find 12021 error,
Description: The PLATE did not move within the correct time.
Cause: A malfunction occurred in one or more of the following components:
• COIL BOARD
• SLOW SCAN AY
Action: 1. Check for an obstruction.
2. Do the checkout for the SLOW SCAN CONTROLLER BOARD A6 and COIL
re: re: re: re: Carestream Classic CR problem
May 27, 2020 11:12
Special-Pricing Available on Medical Displays, Patient Monitors, Recorders, Printers, Media, Ultrasound Machines, and Cameras.
This includes Top Brands such as SONY, BARCO, NDS, NEC, LG, EDAN, EIZO, ELO, FSN, PANASONIC, MITSUBISHI, OLYMPUS, & WIDE.