Agfa DS5302 System Failure
June 16, 2017 05:53
I have an Agfa DS5302 printer that has "system failure" at startup.
You can switch the system on/off even with the error.
I have tried to perform a software re-installation but system does not load the CF card at start-up, it just goes to the same error at the point when its checking the drives.
I don't have an emergency_install so I am using the core software package.
Is there another cause for this error or can some one help with a solution
to rate and post a comment
re: Agfa DS5302 System Failure
June 18, 2017 08:22
Replace themis board.
re: re: Agfa DS5302 System Failure
June 24, 2019 10:39
i am facing same problem with agfa 5302 .whats solution
Hard disk failure maybe but less probable
June 18, 2017 08:23
we are freelance engg. of agfa so plz give me your system in team viewer we are trying to load software
our e mail firstname.lastname@example.org
whatsapp no +918896046247 , +919999393192
June 19, 2017 10:22
It will probably be necessary to replace the themis/tyndall board, if it is the tyndall board (serial number above 12500) and two green leds are flashing or the red led is direct access, replace the tyndall board.
Failure of the SYSTEM ERROR means that the operating system can not be loaded, the items that can cause this failure are:
Themis - Tyndall
Devius - Devils
HD (SCSI or SATA)
Usually the fault is Themis / Tindall.
For equipment with a serial number above 12,500, they will be using the Tyndall card if the 2 green LEDs on the 4 green LEDs that monitor the power supply on the Tyndal card are blinking. The problem is the TYNDALL board, or the red LED is connected directly, you can change the Tyndall board.
June 25, 2017 05:27
The problem consists in bad soldering of CPU. The processor heats up and cooled on which balls it is soldered burst and the contact is lost.
Solution warming up (soldering) of the processor! More than 2 tens printers across Russia were so regenerated already.
June 26, 2017 10:47
re: re: re: Agfa DS5302 System Failure
June 26, 2017 04:21
Try to solder!
re: re: re: re: Agfa DS5302 System Failure
October 11, 2018 08:09
Hello Fedor, Nice to meet you in Dotmed,I like your opinion for System failure i want to try it my DS5302.Pl contact with me by email@example.com Thanks Besuud from Mongolia
I would like to try your method but in which board .Where is i will do it? Please answer me ÐÐ´Ñ Ð¾ÑÐ²ÐµÑÐ°
Craig ,Did you solve your problem? I now have same problem like yours.Could you answer by besuudp@yahoo,com Thanks
October 11, 2018 11:04
Contact me firstname.lastname@example.org
I replaced the Themis/Tyndall boards and performed a software upgrade.
December 04, 2018 05:58
I have the problem with "system failure' too, but the strange thing is that, when I have this error at boot up - status led on front panel, doesn't lights and Themis board was changed one year ago.
I reinstall software with 5.1.0 and have system failure again.
Is it possible to have defect in themis board again ?
I have only D1 and D11 lights(yellow), when turn on the DS5302
December 05, 2018 09:26
I would replace the whole assay i.e. Themis/Tyndal...
Can contact me for help at email@example.com
June 24, 2019 10:38
how to resolve this problem . i am facing same problem
INTUITIVE THINKING • INTUITIVE WORKS • INTUITIVE SOLUTIONS • Contact TIBS today for your GE OEC C-arm Sales, Service or Parts. Parts & systems in stock now. Exchange or sell us your old C-arm. Call 866.499.3966 or email firstname.lastname@example.org.